This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||5 January 2013|
|PDF File Size:||3.29 Mb|
|ePub File Size:||3.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sales and Service Office and are listed below. Installation And Setup Installation and Setup Initial Inspection Inspect the shipping container and the cushioning material for signs of stress.
This function can be auto-coupled. Page 75 Table Creating A Directory b. A total of 3 years of warranty service support.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Retain the shipping materials for future use, as you may wish to ship the analyzer to another location or to Agilent Technologies for service. Make sure there is no disk in the front-panel disk drive. Each softkey More has several characters on it. Place a marker mamual 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and manjal the upper-right corner of the screen.
Page 41 Improving Frequency Accuracy In Case of Difficulty Returning an Analyzer for Service Returning an Analyzer for Service Service Tag When you are returning an analyzer to Agilent Technologies for service, fill out and attach one of the blue service tags provided at the end of this chapter.
Page 18 When you have completed the printer setup for the analyzer, press the front panel key.
The file cannot be placed inside another file. RS This feature not fully implemented; to be implemented in a future release. Activate trace 3 by pressing similar to Figure To increase the accuracy of e4404a frequency reading in the marker annotation, turn on the frequency count function.
Used Agilent EA
Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation The battery is located on the CPU board.
Installation and Setup Turning on the Analyzer for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network recognizes, but that is e4404a yet in use: To access the menu, press Rename 2.
Is the analyzer turned on? For a complete list of spectrum analyzer messages, see the Instrument E4440 and Functional Tests manual.
Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Page 70 In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem.
Agilent Technologies E4440A Getting Started Manual
Select the type of file you want to save. Enter text from picture: File Catalog Figure This symbol is used to indicate the ON position of the power button. EA 3 Hz – Please read it and become familiar with its terms. Put a formatted floppy in the A: Firmware Revision To view the firmware revision of your analyzer, press mxnual Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
EaEaEaEa. Please be as specific as possible about the problem.
Copying a File 1. To access the Save menu, press, 2.